A few weeks ago I mentioned that Photometrics had released a new camera, the Prime95B, featuring a back-illuminated sCMOS sensor with 95% peak QE. I got a chance to play with it last week, and I’m pleased to say that it performs as well as you would expect. We compared it to an iXon 888 EMCCD mounted on our CSU-W1 spinning disk confocal. We had purchased this EMCCD for imaging those samples that were too dim to get good images with the Zyla 4.2 sCMOS camera we also have mounted on the confocal. (You can see a sketch of how everything is configured in this previous post). For testing the Prime95B, we replaced the Zyla 4.2 with the Prime95B, allowing us to directly compare the Prime95B and the iXon 888.
Before I get to the data, however, what performance do we expect? To get a sense of what to expect I wrote a Matlab script that calculates the theoretical performance for a number of different cameras, using their quantum efficiency, read noise, and excess noise factor (for EMCCDs). You can get the script here. You can read more about how to calculate the signal-to-nosie ratio for a camera in this Hamamatsu white paper. Here, I’m ignoring the different pixel sizes of the various cameras by assuming that they all receive the same photon flux per pixel, as if the magnification had been adjusted to produce the same effective pixel size at the sample.